To clean the plunger, remove it from the syringe barrel and gently wipe with a lint-free tissue. Reinsert the plunger into the barrel and pump deionized water or acetone through the needle and syringe. Air dry the syringe for storage.
*Voltage rheostat and vacuum source must be purchased separately
Should a needle become clogged, do not attempt to clean it by forcing liquid or compressed air through the syringe. Excessive pressure will split the glass barrel. Remove the plunger from the syringe before cleaning the needle. Cleaning wires are used to dislodge any foreign material or residue build-up. Use deionized water or acetone to thoroughly rinse the syringe.
Founded in 1983 as a manufacturer's representative, Chrom Tech has grown to a worldwide distributor of chromatography instrumentation and supplies. We are dedicated to providing technical expertise in the chromatography field by recommending products to meet your analytical requirements.